112018May 0 2018 Instrument Cleaning and Sterilization Competency By Alissa Harris / CommentsOff Take the following 2018 Instrument Cleaning and Sterilization Competency Test. You must score 80%. The test can be taken multiple times until this score is reached. Click the Next button to start the test. Items subjected to immediate use steam sterilization must go through the same decontamination procedure as for terminal sterilization. True FalseThis is the only direct measure of the lethality of the sterilization process. Bowie-Dick Test Sterilization Indicator Tape Integrating Indicator Biological IndicatorThe following item is __________________. Unprocessed ProcessedAll staff working in decontamination must wear appropriate PPE, which includes: Fluid resistant gown and gloves Fluid resistant gown, gloves, and face mask with eye protection Gloves Fluid resistant gown, gloves, face mask with eye protection and shoe covers if potential for becoming contaminated or soakedThe following item is ___________________. Unprocessed ProcessedThe Bowie-Dick test below should be interpreted as a Pass FailThe pack on the left is ______________, and the pack on the right is ______________. Processed, Processed Processed, Unprocessed Unprocessed, Processed Unprocessed, UnprocessedTrays may not exceed ________ pounds. 15 20 25 30The following chemical integrator would be interpreted as Unprocessed Processed - Reject Processed - Accept I Don't KnowThe 3M Attest Super Rapid Readout provides results in ______ . 30 minutes 1 hour 2 hours 3 hoursThe biological indicator (BI) should be placed in the sterilizer on the Top shelf near the door, above the drain Top shelf near the door, opposite the drain Bottom shelf near the door, opposite the drain Bottom shelf near the door, over the drainClick Submit Quiz to see your results. A submission email will be sent to the designated OISC Administrator. If you would like a certificate of completion, let Alissa know and one can be printed for you. Time is Up!